I.Normal Installation, Medium/ Large Scale Coriolis Mass Flow Meter,
It is recommended to release and empty the gas that possibly stored in the pipeline of the Coriolis Mass Flow Meters before installation.
II.Inverted Installation, Medium/ Large Scale Coriolis Mass Flow Meter,
It is recommended to use the inverted installation for the measurement of Gas (e.g. steam). Because only in this way, the liquid that could possibly stored in the Coriolis Mass Flow Meters could be released and emptied.
III.Flag- Type Installation, Medium/ Large Scale Coriolis Mass Flow Meter,
Flag- type installation is the installation way that the Coriolis Mass Flow Meter should be vertical to the horizontal line.
The measurement of suspension liquid, and the double-phase liquid of both liquid and solid is recommended to use the flag-type installation. E.g. For the liquid of easy solidification under the change of temperature, and for the liquid of high viscosity, it is required to empty the liquid left in the measuring pipeline after the measurement. The flag-type installation will not influence the measuring accuracy of the any types of liquid measured. But the direction of the measured flowing must be from bottom up.
IV.Tiny Pipe-Size Coriolis Meter Installation
The tiny- size Coriolis Mass Flow Meters’ flow range is very small, while the requirement about the accuracy is high. In this way, then the stable mounting plate and bracket are the must. The installation of mounting plate and Coriolis Mass Flow Meter must be completely flat without any space. If not, please use the level bolt to adjust the balance. The installation of the tiny- size Coriolis Mass Flow Meters is shown as below: