Answer : If coriolis flow meter measure 2 medium (A & B) not incompatible (such as water and oil) , “Percent of A “ is the A medium concentration.
Question: What is “Medium 1 Density” and “Medium 2 Density” used for?
Answer : Two incompatible mediums mix , here customer can input the 2 medium density into the mass flow meter
Question: What is “Density coefficient” used for?
Answer : It is used to calibrate the density if the Coriolis mass flow meter display the density is not very accurate.
Question: What is “pressure correction” and Temperature correction” used for?
-I assume this is a user adjustable calibration trims?
Answer : Pressure correction is reserved for future function update.
Temperature correction is for calibration trims.
Question: Confirm what happens if “Reset” and “Factory Reset” are used?
-Are these the same? -Are calibration constants lost, or just the settable units and span setting etc. lost?
Answer: Not the same.
Factory reset can cause calibration constants lost.
And reset is for the settable units and span setting etc.
Silver Automation Instruments Coriolis Mass Flow Meter catalog: http://www.silverinstruments.com/product/flow-measurement/coriolis-mass-flow-meter.html