Measuring flowrate of various fluids is an integral part of the oil and gas industry. There are several types of devices that are used for this purpose and the Coriolis flow meter is one of them. It has gained popularity over the past years as a tool for measuring the flowrate of liquids and gases for the manufacturers and users of oil and gas skids,also great option used as high viscosity fluids flow meter
Coreolis flow meters incur a higher initial installment cost when compared to other types of flow meters. However, factors like its accuracy and speed of measurement, ability to capture mass readings rather than just volume, and the subsequent lower maintenance costs all work in its favor.
What is Coriolis Flow Meter?
Also known as the Mass Flow Meter or the Inertial Flow Meter, this device functions on the principle of the inertia that is generated by a fluid while flowing through an oscillating tube. This force of inertia results in the twisting of the tube, proportionally to the mass flowrate. The degree of this twist or distortion is captured by sensors in the device and a linear flow signal is given as the output.
Working of a Coriolis Meter
The earlier designs of the Coriolis flow meter
consisted of a single tube that was in the shape of a curve. However, due to the thinness of the wall of this tube, there were problems with vibration. In modern models, this issue has been eliminated by incorporating two tubes that result in a thicker wall. Additionally, the modified design requires a lesser amount of power and yet is high on durability. There are several shapes and styles of tubes available in the market, each suitable for specific applications. All of them are highly precise and low maintenance.
In this design, the tubes are vibrated in opposite directions by using magnetic coils mounted at the inlet and outlet sections of both the tubes. These magnetic coil assemblies function as sensors. When a magnetic field is generated by these magnets, the coils passing through the field create sine waves. In the idle condition when there is no fluid flowing through the tubes, the inlet and outlet sine waves are in phase.
And when there is fluid flow, the tubes twist causing a phase shift in the sine waves. This twisting is proportionate to the mass of the flowing fluid and the sensors capture this phase shift reading. The phase shift is equal to the time difference between the sine waves and this reading gives the mass flowrate.
Pros of Coriolis Flow Meter
There are a number of advantages of the Coriolis Flow meter over other types of flow meters. Some of them are as listed below.
The crucial feature of this type of flow meter is that it is the most accurate of all flow meters available.These flow meters are low maintenance because they do not have any moving parts that can cause wear and tear. They also seldom require calibration. Since they measure mass flow instead of just volume of the fluids, they are suitable for measuring flowrate of materials which are sold by weight and not volume. And in the oil and gas industry value of products like petroleum is measured by heating units (like BTU), thus making the Coriolis flow meter the device of choice to calculate flowrate for such materials. The mass flow meter gives accurate measurements of fluids and is not affected by the changes in temperature, pressure, speed of flow. This makes it excellent for measuring flowrate of gases which are more prone to be affected by the afore-mentioned changes. It is because of this reason that Coriolis flow meters have gained traction in the oil and gas industry.